# 21.5 Null measurements

 Page 1 / 8
• Explain why a null measurement device is more accurate than a standard voltmeter or ammeter.
• Demonstrate how a Wheatstone bridge can be used to accurately calculate the resistance in a circuit.

Standard measurements of voltage and current alter the circuit being measured, introducing uncertainties in the measurements. Voltmeters draw some extra current, whereas ammeters reduce current flow. Null measurements balance voltages so that there is no current flowing through the measuring device and, therefore, no alteration of the circuit being measured.

Null measurements are generally more accurate but are also more complex than the use of standard voltmeters and ammeters, and they still have limits to their precision. In this module, we shall consider a few specific types of null measurements, because they are common and interesting, and they further illuminate principles of electric circuits.

## The potentiometer

Suppose you wish to measure the emf of a battery. Consider what happens if you connect the battery directly to a standard voltmeter as shown in [link] . (Once we note the problems with this measurement, we will examine a null measurement that improves accuracy.) As discussed before, the actual quantity measured is the terminal voltage $V$ , which is related to the emf of the battery by $V=\text{emf}-\text{Ir}$ , where $I$ is the current that flows and $r$ is the internal resistance of the battery.

The emf could be accurately calculated if $r$ were very accurately known, but it is usually not. If the current $I$ could be made zero, then $V=\text{emf}$ , and so emf could be directly measured. However, standard voltmeters need a current to operate; thus, another technique is needed.

A potentiometer    is a null measurement device for measuring potentials (voltages). (See [link] .) A voltage source is connected to a resistor $\mathrm{R,}$ say, a long wire, and passes a constant current through it. There is a steady drop in potential (an $\text{IR}$ drop) along the wire, so that a variable potential can be obtained by making contact at varying locations along the wire.

[link] (b) shows an unknown ${\text{emf}}_{x}$ (represented by script ${E}_{x}$ in the figure) connected in series with a galvanometer. Note that ${\text{emf}}_{x}$ opposes the other voltage source. The location of the contact point (see the arrow on the drawing) is adjusted until the galvanometer reads zero. When the galvanometer reads zero, ${\text{emf}}_{x}={\text{IR}}_{x}$ , where ${R}_{x}$ is the resistance of the section of wire up to the contact point. Since no current flows through the galvanometer, none flows through the unknown emf, and so ${\text{emf}}_{x}$ is directly sensed.

Now, a very precisely known standard ${\text{emf}}_{s}$ is substituted for ${\text{emf}}_{x}$ , and the contact point is adjusted until the galvanometer again reads zero, so that ${\text{emf}}_{s}={\text{IR}}_{s}$ . In both cases, no current passes through the galvanometer, and so the current $I$ through the long wire is the same. Upon taking the ratio $\frac{{\text{emf}}_{x}}{{\text{emf}}_{s}}$ , $I$ cancels, giving

what is a good calculator for all algebra; would a Casio fx 260 work with all algebra equations? please name the cheapest, thanks.
a perfect square v²+2v+_
kkk nice
algebra 2 Inequalities:If equation 2 = 0 it is an open set?
or infinite solutions?
Kim
y=10×
if |A| not equal to 0 and order of A is n prove that adj (adj A = |A|
rolling four fair dice and getting an even number an all four dice
Kristine 2*2*2=8
Differences Between Laspeyres and Paasche Indices
No. 7x -4y is simplified from 4x + (3y + 3x) -7y
is it 3×y ?
J, combine like terms 7x-4y
im not good at math so would this help me
how did I we'll learn this
f(x)= 2|x+5| find f(-6)
Need to simplify the expresin. 3/7 (x+y)-1/7 (x-1)=
. After 3 months on a diet, Lisa had lost 12% of her original weight. She lost 21 pounds. What was Lisa's original weight?
what is nanomaterials​ and their applications of sensors.
what is nano technology
preparation of nanomaterial
Yes, Nanotechnology has a very fast field of applications and their is always something new to do with it...
can nanotechnology change the direction of the face of the world
At high concentrations (>0.01 M), the relation between absorptivity coefficient and absorbance is no longer linear. This is due to the electrostatic interactions between the quantum dots in close proximity. If the concentration of the solution is high, another effect that is seen is the scattering of light from the large number of quantum dots. This assumption only works at low concentrations of the analyte. Presence of stray light.
the Beer law works very well for dilute solutions but fails for very high concentrations. why?
how did you get the value of 2000N.What calculations are needed to arrive at it
Got questions? Join the online conversation and get instant answers!